(1) Backboard can ascend 80 degree and footboard can ascend 40 degree.
(2) Ascending and descending scope of bed surface: 325 cm.
(1) One set of grating guardrail.
(2) One piece of two-section drop frame.
(3) One set of 5" brake wheel of dust-proof shield.
(1) The design concept of children medical bed mainly focuses on the safety of victims, operational demands and convenience of nursing personnel.
(2) Grating guardrails are equipped on the both sides of bed, which are 60cm in height and can prevent children from climbing or falling while turning over. And the increasing and decreasing switch of guardrail is safe switch, which can prevent children from danger while playing and turning on. The whole bed is equipped with ducts with rounded upper covers except processed steel tube of acute angle, and all screws are the cup head for avoiding scratch.
(3) The bed was designed as three-rocker type based on its function, besides the backboard and footboard can be lifted, the bed can receive inbuilt ascending and descending or; and the bed surface can be operated conveniently for nursing personnel while ascending, and the effective maximal height of guardrail would be achieved while descending.(If apply the ascending and descending method of common medical bed, the guardrail with a height of 60cm would result in that it cannot be operated for higher bed).
1. Specifications: (1) L2000xW950mmxH1400mm. (2) Height of bed surface: H585~910mm. (3) Dimension of bed surface: 1880x780mm.
2. Material: the bed is made of steel and coated with power baking varnish. Bed panel is made of ABS and molded.